http://www.imedison.comAgency ContactLianhua NanPhone: 86 - 0631 - 5337707/+86 - 13306316804Fax: 86 -
Qujiahe Industrial Park,Yangting Town,Huancui DistrictWEIHAI, shandongChina, 264400Web Page Link: https://www.toboc.com/weihaimedisonmedicalequipmentcoltd.aspxD&B D-U-N-S Number: 008606310
Disposable Circumcision Stapler is mainly used for circumcision surgery and is combing cutting and suturing operation. The technology is a huge improvement in urological surgery which makes complicated circumcision safe, simple and easy to operate. At the same time, the technology solves the problem of swelling and pain caused by circumcision ring devices. The product is disposable to avoid cross infection.Operating Principle:Disposable Circumcision Stapler is new type circumcision device using for treatment of foreskin or phimosis and combining cutting and suturing surgery operation. The operationg principle is similar to the Procedure for Prolapse and Haemorrhoids(PPH) operating principle. Through mechanical transmission, circumcision stapler staples foreskin needed to be cut and sutured with the pre-set suture nail. The suture nail will be transformed into& 160& 160& 160 type when be resisited by nail resistance base and cut the redundant foreskin.Advantages:Slight pain, no cutting deformation, rapid recoverySave time of suture and bleeding stoppingAvoid discomfort caused by stitch residue of suture ligation.Instructions for Use:Preoperative preparation:Skin preparationlocal anesthesia or general anesthesia (Sacral anesthesia) Deal with complicaiton firstly: phimosis patient: need to dilate preputial orifice or use scissors to cut prepuce if necessary to let bell-shaped nail resistance base be put in. Adhension patients need to be isolated. Use forceps to clamp hood foreskin (frenulum) in the movement point between the inner and outer prepuce. Use two or four pairs of forcep to pull the foreskin in other positions. Keep frenulum slighty loose to ensure frenulum can not be over cut. Insert bell-shaped nail resistance base into foreskin. Make concave shape base cover of nail resistance assembly(bell jar) cover the glans.Put the glans into the bell jar in the position of frenulum. Incline the vertical axis of the bell-shaped nail holder at an angle to the vertical axis of the penis to keep edge of bell base and coronary sulcus in the same direction to make the cutting position right.Fix the foreskin to the rod with suture. Re-check and make sure the to be cut part and the foreskin fixed to the rod completely.Confirm the cutting position and make sure the frelunulum in slightly position. Cover the device, turn up and tighten adjusting knob to the end face of the pull rod, which is flat or slightly protruding (do not exceed 0.5mm). The two "ears" of the adjusting knob should avoid the direction of the handlebars so that there will be no conflict when the handlebars are pressed. Take out the safety pin and hold the front 2/3 of the handle in one hand for pressing hardly.The other hand holds 1/3 of the handle to stablize the handle. Presse handle & 160simultaneously or successively,& 160 the foreskin will be cut and sutured at the same time.Release handle a few seconds later. Use hands to release handle if the handle can not release itself. Hold device with one hand, and turns the adjusting knob to 5-10 mm with the other hand.Loosen the cutting position by pushing the adjusting knob forward. Observe wether the to be cut foreskin is completely cut offIf not, the little foreskin can be trimmed with blades or scissors.Wrap the surgical part with dry gauze Immediately.Press to hemostasis with fingers separately on average for 5-10 minutes. (The extrusion process during cutting and suturing can cause tissue bleeding) Loosen and check the suturing condition (the process can be omitted after skilled surgery)Pressing and dressing can be performed directly after cutting.If rare punctate ejection happens, doctor must sew it to stop bleeding.
View All Products posted by
Weihai Medison Medical Equipment Co., Ltd,