http://www.ynoilpurifier.comExport Departmentlily aaaaaaaaaaaPhone: +86 - 23 - 62940030/+86 - 15909333942Fax: +86 - 23 - 86-23-62940030
No. 9-10, 23 Nanping West Road, Nanping Street, Nan'an District, Chongqing, Chinachongqing, chongqingChina, 400060Web Page Link: http://www.toboc.com/chongqinggoldmechanicalelectricalequipmentc
chongqing gold mechanical & electrical equipment co.,ltd
no. 9-10, 23 nanping west road, nanping street, nanan district, chongqing, china
the instrument is designed and made as per the national standard gb/t4507 “test methods for softening po int of petroleum asphalt”, and t0606 “asphalt softening point test (ring and ball methods) in the industry standard jtj052 specification and test methods of bitumen and bituminous mixture for highway engineering. it is suitable to determine softening poin t of petroleum asphalt, coal pitch, liquid petroleum asphalt, and other asphalts. it is the desired products for asphalt manufacturing companies, highway and bridge construction companies, colleges and universities, and scientific research institutes.
c omputer control; linearity heating; automatic stirring. the test data is accurate and reliable. it can show test data automatically after a test. it is in accord with the national standard gb/t4507.
1. power supply: 220v 7;10 %, 50 hz;
2. ambient condition: room temperature should be less than 35 ?, and keep stable without any air convection.
3. measurement range: +32~150?, ±1.0?;
4. stirrer: the stirring speed can be continuously adjusted.
5. heating power: 800 w;
6 . heating medium: glycerin;
7. cubage of beaker: 1000 ml;
8. relative humidity: =85%;
9. total power consumption: not more than 850 w;
View All Products posted by
Chongqing Gold Mechanical & Electrical Equipment C